Thanks for your infomation. Message 8 of My question is how can I read data from the instrument? Sign up for a new account in our community. That error is a VISA timeout, so it means your instrument is not responding and probably not getting your command string, either. I’ve found it to be a useful troubleshooting tool. Is it stand alone operate?
|Date Added:||16 November 2008|
|File Size:||55.64 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Make sure under Serial settings that the labviea settings match the ones in your driver. Simple code for HPA. But using LabView still not work, what missing.
Create an account or sign in to comment You need to be a member in order to leave a comment Create an account Sign up for a new account in our community. I tried hp34401a labview have same configuration as in the hp34401a labview7, Even, 1, Hardware both sides but same error. hp34401a labview
Register a new account. If it does, then you are probably communicating correctly with the instrument – if not, then you first need to figure out how establish communication before you try hp344011a make a measurement. It is nine to meet you. Message 8 of Did you read that? That error is a VISA timeout, so it means your instrument is not responding and probably hp34401a labview getting your command hp34401a labview, either.
Hi, I am new to this forum and I need little help in LV.
I checked “Find Instument drivers” and nothing under connected inst. Now I got error no.
Agilent Technologies / Keysight Technologies 34401A
Auto-suggest helps you quickly narrow down your search results by hp34401a labview possible matches as you type. Most Active Software Boards: Sign in Already have an albview There you hp34401a labview get automatically installed an instrument driver into your functions palette. My question is how can I read data from the instrument? It seem beginning to work.
That is really good but problem hp34401a labview it gives same wrong readings ,97 or 33,98 orBut hp34401a labview Aglient IO control has see the device.
The interface cable must also have the proper connector on each end ” Mark.
Message 3 of Started by MouradJanuary 21, For this issue, how can I hp34401a labview it. Sign up for a new hp34401a labview in our community. Also, your cable may still be incorrect – have you tested pin-to-pin to make sure it matches the diagrams in the user manual?
I think this can only be done from the ‘s menus – not programatically.
TEGAM Instrument Driver Readme
Can you post your code for us to take a look at? Message 2 of Posted January 27, Message hp34401a labview of You are right hp34401a labview the cable, I hp34401w it and it was not totally crossover one.
I am new to this forum and I need little help in LV. Posted January 22,